1980s Officer's USSR Military Geiger Counter Dosimeter ID-1 Soviet For Sale
This item has been shown 61 times.
1980s Officer's USSR Military Geiger Counter Dosimeter ID-1 Soviet :
Description please see on YouTube!description:Set the ID-1 dosimeters F16 059069- A set of 10 dosimeters ;- Charger ZD-6 #F16 046721;- Form;- Technical description and user manual;- Case;
The dosimeter ID-1 is made in the form of a pen (pencil) and consists of an ionization chamber, an electroscope, a condenser, a microscope and a contact group in an aluminum casing.Extract from the instruction, the dose meter ID-1 provides measurement of absorbed doses of gamma-neutron radiation in the range from 20 to 500 rad.When exposed to ionizing radiation in the volume of a charged ionization chamber, cylindrical electrodes which are made of tissue-equivalent plastic, a current appears that reduces the potential of the chamber and the associated capacitor. At the same time, the filament of the electroscope to which the potential of the capacitor is applied is deflected and its deflection is measured with a microscope on a scale calibrated in radars. The scale has 25 divisions, the division price is 20 rad. To ensure the linearity of the scale, the charging potential of the ionization chamber is selected within the limits of 180-250 V.The contact group of the dosimeter is charged with a charging device; After the charge, the contact group protects the dosimeter from discharge.
Complete kit dosimeters ID-1 shakeproof and durable during transportation;The dosimeter is sealed;Overall dimensions of the set in the packing, the charger and the dosimeter does not exceed the following values: a) ID-1 184 X 102 X 142mm; b) charger ZD-6 105 X 37 X115mm; c) the dosimeter holder: diameter - 19mm length - 128.5mm; Mass kit in packing, the charger and the dosimeter does not exceed the following values: a) set the ID-1 in the installation - 1500g; b) charger ZD-6 (without power supply) -500g; c) the dosimeter - 40g.Specifications instrumentation kit dosimeters ID-1:Count exposure doses measured on a scale, and located inside the dosimeter calibrated in X-rays;The main error of measurement under irradiation source Co60 in normal conditions - does not exceed ± 20% of full scale Note. Normal conditions are: - Temperature - + 20 ° C ± 5 ° C; - Atmospheric pressure - 750mm.rt.st ± 30mm.rt.st .;. - Relative humidity - 65% ± 15%;
You get what you
ANY DEFECTS NOT
MENTIONED BUT VISIBLE IN THE PICTURES WILL BE CONSIDERED AS DESCRIBED
Payment: I accept PayPal.
PLEASE DO NOT FORGET TO GIVE YOUR response.
YOUR response IS
BY AIRMAIL !
I am always sending
a parcel with a track number !
You can track
shipment online - trackitonline.org
Thanks in advance
if your item is delayed in the mail, do not rush to give negative response or
open a class="MsoNormal">Write an e-mail
message - I can solve any problem.
I AM ALSO NOT RESPONSIBLE FOR ANY CUSTOMS
DUTIES,TAXES,ETC IMPOSED BY YOUR COUNTRY.
Professhional Ftlab Ftc-001 Smartlab Smart Temperature Humidity Checker
As1392 Digital Lcd Emf Tester Electromagnetic Radiation Detector Counter E0v3
Gxg-1987 Gm3120 Lcd Digital Electromagnetic Radiation Detector Dosimeter Tester
Kkmoon Emf Meter Handheld Mini Digital Lcd Emf Tester Electromagnetic Field Radi