N1201SA UV RF Vector Impedance ANT SWR Antenna Analyzer Meter Tester 140MHz - 2. For Sale
This item has been shown 23 times.
N1201SA UV RF Vector Impedance ANT SWR Antenna Analyzer Meter Tester 140MHz - 2.:
Description:N1201SA series is the handheld radio frequency vector impedance measurement analysis instrument, easy to use, simple operation. Built-in high capacity lithium ion battery for mobile and outdoor use.Shutdown automatically save settings parameter without human intervention, after starting up, interface settings completely the same as the power off before, use very worry.Can be used to test the antenna, the imppedance of the radio frequency devices.As RF impedance measuring need calibration, so you need to purchae separately currency:140MHz~2700MHzActual currency :137.5~2700MHzStepped frequency: 1kHzDisplay:2.4" TFT trdpResolution ratio:320 x 240(QVGA)Battery capacity: 2000mAH(7.4Wh)Power consumption: <1.5WCharge current: 400mACharge port:USBAuto power off can be set 5-60minutes.Measured parameters:Resistence, Reactance, Standing wave, S11Resolution ratio: 4 numberFrequency accuracy: about range:Impedance:0.1~1000Standing temperature: 0~40℃Atmospheric pressure: 860hPa~1060hPaThis instrument has fourinterfaces:single point measurement, scanning, system information, correction and calibrationThe default boot into single-point measurement interface.
Note:For the convenience of the user, each time calibration correction options will automatically change to the user calibration. Meanwhile, in order to ensure the integrity of the calibration, the calibration process, the automatic shutdown function fails, the consistent boot.Startup and shutdownHold down the CTRL key off press the power switch button.The boot state, long press (2S above) the switch button, after the release, the instrument will shut down.The boot state, long press (2S above) the switch button, after the release, the instrument will shut down.Recommend the use1. In order to ensure the accuracy of measurement and calibration, preheat the instrument at least 15 minutes.2. Try to use good quality RF cable and connectors.3. Calibrated cable marked to the next may not need to calibrate and direct use.4. Test cable process affect the measurement accuracy, it is recommended that, where possible, do not choose too long test cable.5. Measurement and calibration before, make sure you are not in a strong wireless signal, so as to avoid interference caused by inaccurate measurements.6. If the impedance measurements themselves do not understand, you can refer to a vector network analyzer instructions and tutorials. This instrument is equivalent to a single-port vector network analysis.
Package included:1 x N1201SA ANT SWR Antenna Analyzer(English Version)
Ridgid 42348 Seesnake Max Rm200 Inspection Camera System With Transport Kit
Mastech Multifunctional Stud Metal Detector Ac Voltage Scanner
Dajac - Expert Io 1000 - Surplus
Dajac - Expert Io 1000 - Surplus Control Automation